What is Ankle Brachial Index?
The Ankle Brachial Index is a standard and common method of physiologic assessment of vascular pathology in the lower limbs in a fast and simple manner.
Systolic pressures are measured in each of the right and left brachials, as well as at the level of the right and left ankles. ABI is then determined as the ratio between the higher brachial systolic pressure and each of the ankle pressures. Thus, a separate ABI value is obtained for the Right side and the Left sides.
Unlike ultrasound imaging modalities which provide anatomical information, ABI assessment provides a direct indication of the physiological severity of the vascular pathology. ABI is often used for fast and simple screening of the lower limbs, prior to determining if ultrasound imaging is required.
How to Measure Ankle Brachial Index
Measuring the Systolic blood pressure is simple. Once a pressure cuff is wrapped around the measured site, a distal Doppler or Photoplethysmograph (PPG) waveform is obtained. The cuff is then inflated until the distal signal disappears as a result of the total occlusion. The pressure cuff is then deflated in a controlled manner, and the pressure at which the distal signal re-appears is determined as the systolic pressure.
Using Doppler to measure the distal waveform is the traditional and clinically accepted method, although PPG sensors are also used in specific conditions or locations.
Using the Falcon for ABI measurements
The Falcon allows very simple and fast diagnosis of ABI. Simply place 4 color-coded pressure cuffs on each of the right/left brachial and ankle sites, select the ABI protocol, and you are ready to go. The test can be completed in just a couple of minutes.
The Falcon allows you to select from a wide variety of Doppler probes and PPG sensors for best outcome. All probes, sensors and pressure tubes are color coded to allow optimal user interface and correlation with the dedicated software. The target inflation pressure, as well as the deflate rate and various other parameters can be customized by the user for optimal measurements.
Numerous other Falcon features and options, such as automatic cuff inflation once a signal is identified, or simultaneous measurements, display of contralateral results, and many more options, are designed to simplify the use of the Falcon ABI physiologic system in a fast and efficient way.
Once the distal Doppler or PPG waveform reappears during the deflation phase of the pressure cuff, the measurement can be stopped, and the Falcon places a cursor to mark a potential systolic pressure. The user may move the cursor to a different location, and the systolic pressure is changed accordingly. R/L ABI are calculated automatically by the Falcon and presented clearly on the screen and in the report layout.
|ABI ≥ 1.4||Non compressible arteries|
|1 ≤ ABI < 1.4||Normal|
|0.9 ≤ ABI < 1||Borderline|
|ABI ≤ 0.9||Abnormal|
Overview of Peripheral Arterial Disease of the Lower Extremity, Ali F. AbuRahma and John E. Campbell, Noninvasive Vascular Diagnosis, A.F. AbuRahma (ed.), Springer International Publishing AG 2017, Ch 21, pp 291-318
2016 AHA/ACC Guideline on the Management of Patients With Lower Extremity Peripheral Artery Disease; A Report of the American College of Cardiology/American Heart Association Task Force on Clinical Practice Guidelines; Journal of the American College of Cardiology, Vol. 69, No. 11, 2017
Peripheral vascular disease assessment in the lower limb: a review of current and emerging non‑invasive diagnostic methods, Shabani Varaki et al, BioMed Eng OnLine (2018) 17:61
2017 ESC Guidelines on the Diagnosis and Treatment of Peripheral Arterial Diseases, in collaboration with the European Society for Vascular Surgery (ESVS), Victor Aboyans et al., European Heart Journal (2017) 00, 1–60
The Ability of the Toe-Brachial Index to Predict the Outcome of Treadmill Exercise Testing in Patients with a Normal Resting Ankle-Brachial Index, Christian Høyer et al, Annals of Vascular Surgery, Volume 64, April 2020, Pages 263-269
Association of cardiovascular autonomic dysfunction with peripheral arterial stiffness in patients with type 1 diabetes, Lía Nattero-Chávez et al, The Journal of Clinical Endocrinology & Metabolism
Measurement and Interpretation of the Ankle-Brachial Index, Victor Aboyans et al, Circulation. 2012;126:2890–2909
The effect of percutaneous transluminal angioplasty of superficial femoral artery on pulse wave features, Mikko Peltokangas et al, Computers in Biology and Medicine 96 (2018) 274–282
Disclaimer of Information & ContentThe content of Viasonix Ltd. website is for information only, not advice or guarantee of outcome. Information is gathered and shared from reputable sources; however, Viasonix Ltd. Management is not responsible for errors or omissions in reporting or explanation. No individuals, including those under our active care, should use the information, resources or tools contained within this self-diagnosis or self-treat any health-related condition. Viasonix Ltd. Management gives no assurance or warranty regarding the accuracy, timeliness or applicability or the content.